template files (this info may be
outdated now that P2PP doesn't just run on kila):
-- See later notes on P2PP below.
Log in as instmgr@killa. The files used by P2PP are in directories:
~/instruments/EFOSC (used by engineering/testing)
/EFOSC2 (used by P2PP)
*.tsf are the template signature files. When making changes to the.tsf
1) delete index.EFOSC2
2) Run p2pp from the instmgr account -- this will generate a new index
file. (See README.install.new-IP written by Oli for instructions to
archive to Garching the new.tsf files).
OBs with P2PP -- some notes
Be sure to hit return or tab after making any changes so that the value
normal readout -- 1 amplifier
fast readout -- 2 amplifiers
There is _no_gain_ from windowing the CCD readout -- don't advice it.
The binning should be 2x2 unless seeing < 0.6"
"Return to origin" = return after the entire sequence.
offsets: each is relative to the one before. These are
_telescope_offsets_ (+1" in RA goes 1" E, +1" in DEC goes 1" N.).
"Combined Offset" = move telescope and guide probe simultaneously, so
that guiding is not lost, and the offset is accurate.
The observing sequence is always Point, Offset, Observe.
"Exposures" = number of offsets, cycled through the list of offsets.
"List of exposure time" = exposure time of each offset (cycled through
list of offsets). For example, if number of exposures = number of
offsets = 5, and list of exposure times = t1, t2 and t3, then the
Exposure 1 2 3 4 5
Offset 1 2 3 4 5
Exp time t1 t2 t3 t4 t5
For coronograph, in the afternoon, should have taken an image with the
mask in. From this image, measure the (x,y) of the mask centre. This
goes to the "move tp pixel" field in the acquistion template.
4) Polarimetry (imaging)
If "continuous rotation" = F -- use list of fixed angles
= T -- continuous rotation
"Move half wave plate in" = T
Exposure time -- it is for each angle.
= N*T where N os the number of postions.
In the afternoon, take image of the Wallaston mask with and without
prism to find out the centre of the gaps. Measure the y positions of
the cetres of the strips where the light is coming through when the
prism is inserted. When using "Move to Pixel" the object must be in
one of these (y) positions.
Let observer decide where to put object by bringing them to the
telescope in the afternoon.
Putting Wallaston Prism into the light path shifts the image location
by about 1 pixel -- beware! (and warn observer).
MUST put Wallaston prism in the FILTER WHEEL -- grism has to go to
grism wheel. Observer must give enough warning for the prism to be
Because the masks mask objects in one direction, can only use one of
the parallel or perpendicular prisms (I can't rememberthe definition
of parallel or perpendicular -- probably the perpendicular one) MUST
CHECK SETUP IN THE AFTERNOON.
6) Standard stars OBs
OB for both imaging and long-slit spectroscopy are in the impex-stdcal
directory. It can only be modified by e3p6ops, and it is under
login to 360cat@aylla
> setenv DISPLAY etc
Macro - Run Macro
NRlists -- all night reports
All_problems -- every problem solved or not
Pending_problems -- problems not closed or solved.
Sciop daily meeting 1.30pm at Library
Tuesday weekly meeting at 5pm at Ritz
Thursday La Silla Action Point meeting, 2pm
Typical daily schedule:
1:30pm Sciop meeting
2:00pm La Silla Action Point Meeting (Thursdays only)
2:30pm Meet visiting astronomer. Assign p2pp number, userame.
4:00pm Bring astronomer of night to telescope for calibrations.
5:00pm Weekly team meeting (Tues only)
Sunset: twilight calibrations, start of observations
10:00pm Meet astronomer of the following night for
> setenv DISPLAY etc
> exmh -- checks emails sent to 360cat account. Must
forward emails to relavant people.
360v1,360v2,360v3 on kila
lasilla@kila -- La Silla Account
efomgr@w3p6ins -- EFOSC magangment. EFOSC2 instrument scientist in
charge of this account.
efosc@w3p6ins -- Instrument account.
guest@wp36drl -- guest data reductions/cdburning
1) Close mirror covers,fans.
2) Go upstairs to close hatch
3) Park telescope: in TCS press Engineering_Presets -- Zenith
4) Go upstairs, goto the TCS rack.
- Turn telescope offline
- Turn off the alpha, delta drive (put in the "pins"). Keep the
button pressed until red
- Turn off oil.
5) Close wind shield, close doors.
To turn on telescope
Reverse the above procedures
1) Turn on oil on TCS rack
2) Start alpha and delta drives (press button until green)
3) Press the reset button on bottom left rack
5) Press reset again
6) Go back to control roo, to open mirror cover etc.
> export R_DISPOSE = "lp -d lpt3p6 %s" to
print directly to printer
> export R_DISPOSE = "mv %s /tmp/saoimage.ps" to dump to
astro@w3p6off -- This should be where visitors reduce data.
guest@w3p6drl -- Old linux data reduction machine. Not supported.
Observer can reduce data here if he/she likes. Also for CD backups.
efosc@w3p6ins -- Instrument machine where the logfile of the
night and the raw data can be found. Do not enter unless necessary
Data flow chart
In order that the data is transferred from the dhs machine to the off
machine, the data subscriber needs to be running.
To run data subscriber on the off machine:
- enter ID name
- to set up preference choose "Stop", "Config"
Set up preference:
"From date" = yyyy-mm-dd
choose "Rename to keyword 'Name on INS WS'"
No header translation.
No post command.
Press "Start Raw".
If data subscriber does not work, try to do the following:
Open xterm. Log in as efosc on ins machine.
click "dhs + obs logs" (kill everythin on off machine beforehand")
Choose "DFSapps", "Start" (or stop), "vcsolac"
Don't do this unless really necessary.
To see if volac is running:
"RTAP" - "RTAPPERFMON".
Data flow-- user backups
Personal backup -- recommended by ESO. User should backup data in the
morning. Data are on w3p6off (astro account) in directory
- Note that in the morning, the calibration files after 12:00 UT go to
the directory of the following night-- remember to move them to a new
directory before night begins, or files will be overwritten.
- Remember to reset the data subscriber (proposal ID, observer name)
before each run.
- Remember to move files on the drl pc to new directory before each
night begins, or at the end of each night, or they will be overwritten.
These are kept at efosc@w3p6ins:/home/efosc/EFOSC_YYYY-MM-DD.log, and
To display log files in real time:
1) On w3p6ins
select "dhs & obs logs" (beware of opening windows twice -- check
if it is already running)
"GP Centre position" to put GP in optical path.
"Guide probe focus" absolute ~ 11,100
Check that everything indicates "Online" (Except mirrir vent says "not
init" but it always is.You can hear the noise of the mirror
ventilation, and also the comments at the bottom of the display
indicates its status).
If not online, "Substate" - "Tracking" will say errors.
If telescope is not trackking, go to "TCS status" window, to look at
"Telescope State" that the "RA axis" and "DEC axis" are "ONLINE".
not "ONLINE", do "OFF" "INIT" "STANDBY" "ONLINE" one after the other.
One can see problems in "Log Mon + PrPrep" message window.
Interlock Panel -
everything must say ok (green).
There will be messages in the l3p6ra-, l3p6dec-, l3p6ada- machine
windows. These messages show what's happening with the LCU's
Control Units". These are interfaces between software and hardware.
Under telescope control system "Dome mode" =auto or manual.
When taking observations make sure that "dome mode" is set to "auto".
Note that under "Engineering presets" option at the top of the window,
one can choose between "zenith", "flatfield" or "pole". The dome will
move to the optimum position automatically, no matter whether the
DomeMode is auto or manual.
on grism alignment:
Object inclined, skylines horizontal -- grism is inclined
Object vertical, skylines inclined -- slit is inclined.
Kurster's Midas Script for
quickly extracting spectra:
change cuts (remember to separate numbers by ",")
Mark 6 points for sky, object, sky.
Wavelength calibration (0, 1 or 2)?
1 -- Give the HeAr frame (eg Arc.fits)
Order of polynomial (5 ok)
Mark lines (left mouse button to mark, other to exit)
need 6 lines for 5th order polynomial
check if the fit looks ok
save wavelength solution table (eg Gr1)
2 -- give the name of previously obtained solution table (eg Gr1)
Give root filename of resulting.fits and.ps files.
useful Mida commands:
> set/graph pmode=1
> plot filename (this is for a.bdf file)
> set/graph yaxis=-200,2000 (change limits)
> set/graph colo=2 (change colour)
> set/graph pmode=0 font=1 yaxis=-200,2000 (another example)
> over filename1 (overplots with file filename1)
To check the wavelength definition/pixel scale of the
> read/desc file.fits
To execute unix command, preceed by a "$".
characteristics on web:
When CCD tests are done, an email will be sent to 360cat, and a file
will be dumped in meli (now epu) under t3p6web. So update must be run
telnet meli (now epu) as t3p6web
/home/t3p6web/bin/update is a script to update the CCD characteristic
plots in the web pages. This should be done every week.
> update <no options> -- prints out the options
> update CCD_number ID_mode
CCD_number -- 40 is the usual CCD used
ID_mode -- 42 is the most used mode (2x2 binning normal readout)
/home/t3p6web/ -- top directory of web page
/home/t3p6web/360cat -- 360cat page
/home/t3p6web/360cat/efosc -- efosc web page
** Important-- mark which side is up or down and mark the field name on
Remove tapes, keep mask flat on table.
Insert into the press, and cut plate.
Put mask into telescope. Select grism and mos, no filter, take internal
exposure with quartz lamp to get the slit positions.
How to reset the punching machine
Turn off CAMARC (bottom box) using orange button at bottom.
Turn on CAMARC.
Use remote control:
Turn to position 2. Use "joystick" to illuminate "Unlock" and "LLIM"
(lower limit) and "Park" (i.e. all red LEDs of these positions are on).
Reset LCU (white button on the middle rack).
Here is the status of the punch heads and tools. There has been a lot
of confusions. Do not confuse the punch tool (which is the machanism
which houses the punch head), and the punch head itself.
(old) Punch 2: Ernesto has put a punch head of 1.75"x11" to the old
punch tool #2. So the original EMMI label on it is wrong. Now we have
renamed this punch tool + head combination as "punch 6", and removed
the old label. The EMMI label is now covered with black tape to avoid
Punch 5: This corresponds to 1.35"x11" on EFOSC. This is labelled.
(old) punch 6: both the punch head and the punch tool are bad. The
punch tool has a broken mechanism. Ernesto is going to contact
Garching about a replacement or instructions to repare it (Nov 2001).
The punches now correspond with the specifications in the old MOS
manual and with those in the program xm.
New punch 6 = 1.75"x11" = punch #6 or #7 in xm =
0.249x1.6mm in mos manual
Punch 5 = 1.35"x11" = punch #4, 5, or 8 in xm =
0.19x1.6mm in mos manual
Please note table 1.2 in the old EFOSC manual is wrong and this will be
updated in the 2003 version.
EMMI 1.02" == EFOSC 1.35"
EMMI punch 1 (0.15mm x 1.0 mm) = EFOSC 1.06" x 7.1". Instrument
/vlt/insroot/SYSTEM/COMMON/SEQUENCES/*.seq -- sequence/scripts
/vlt/insroot/SYSTEM/COMMON/SEQUENCES/*.tsf -- definition of
/vlt/introot/lib/libEFOSC.tcl/ -- directory of library
files of EFOSC
-- script library of subroutines
EFOSClib.tcl -- file with all the subroutines
/vlt/insroot/SYSTEM/COMMON/MIDAS/PROCS/ -- comtains MIDAS routines
(names are EXT*.prg)
/vlt/insroot/SYSTEM/COMMON/OBD/efosc/ -- these are files that can
be sent to BOB if P2PP fails.
instmgr@w3p6dhs (this is account to help organise P2PP):
~/instruments/EFOSC2/*.tsf -- template signature files
-- these are the files that P2PP on the dhs machine sees.
-- when updating.tsf files, must put copies in kila, dhs as instmgr,
ins as efomgr, and also to the web link.
Under ~/instruments/ directory, keep EFOSC2.isf which contains the
To update/make changes to.tsf files (note this section is old-- new
procedures available via CCS-- see later).
1) Delete summary.idx (important)
2) Make the changes to templates.tsf
3) Restart p2pp on dhs as instrument manager (instmgr)
Keep old copies in ~/instruments/
eg tar cvf efosc2.tar EFOSC2* to make backup of the directory before
making any changes.
Afterwards update the copies on kila and ins as well. Also must update
the ESO archive.
Login to instmgr@kila
cd ~/InstrumentPackageArchive/ -- where the old templates (tsf files)
are kept (eg EFOSC2_IP_20001005.tar
See README.install_new_ip under ~/instruments, can find
read: VLT Programming Standard, Configuratio Management Module user
-- This is the correct method.
After a module is updated and archived, it is ready for installation:
Login is efomgr@w3p6ins
> cd ~/CURRENTLY_MOD/
> cmmCopy efiotm -- this
downloads the module
> cd ~/CURENTLY_MOD/efiotm/src
> make clean all man install
copy -- This compiles and install
the modules on the INS machine.
On the ins machine
> ~/CURRENTLY_MOD/efiotm/bin/efiotmInstall -- to install
obd.seq and tsf files to $INS_ROOT.
this removes redundant obds.
Check also files in /vlt/instoot/SYSTEM/COMMON/SEQUENCES/ to see if
they are up to date or need to be deleted.
Updating.tsf and.isf files on w3p6dhs
On w3p6ins as efomgr, run efiosfReplication under
efiosfReplication makes copy of.tsf &.isf files to the dhs
machine. Make sure that p2pp is not running on the dhs machine when
Before running this, go to dhs machine as instmgr (can enter directly
using rlogin from efomgr@w3p6ins). Remove old.tsf files under
~/instruments/EFOSC2/*.tsf, or move them somewhere else. The new.tsf
files will be put under the EFOSC2 directory when the replication
script is run.
Updating P2PP on w3p6dhs
on the dhs machine:
as instmgr, delete ~/instruments/EFOSC2/summary.idx
run p2pp as instmgr to create new summary file.
Making the new instrument package
on dhs machine,
> cd ~/instruments/.
> tar -cvf EFOSC2_IP_YYYYMMDD.tar EFOSC* (this takes the
EFOSC2.isf file as well as the EFOSC2 dir).
Copy EFOSC2_IP_YYYYMMDD.tar to
1) EFOSC web page. Login to t3p6web@meli and put it under efosc/docs.
2) On your computer at the same location.
3) On both your computer and meli, update all references to the IP in
the html files.
4) Email amchavan, mperon, usg, email@example.com.
test some of the Fiera/TCL commands
On the ins machine:
ccs command panel (ccsei Msg)
To test script
> seqWish &
seq_ccsInit (do this first to use message system of VLT)
msgSend -- sends a command.
eg msgSend -v wefosc fcdconCI_efosc STATE " "
seqWish > source /vlt/introot/lib/libEFOSC.tcl/EFOSClib.tcl